- Poster presentation
- Open Access
Clinically relevant skull models and optical measurement method to evaluate programmable hydrocephalus valve tool kit usability
© Luedtke et al. 2015
- Published: 18 September 2015
- Usability Testing
- Shunt Valve
- Time Valve
- Skull Model
- Programmable Valve
Five simulated skull models were created to simulate typical clinical conditions experienced while programming implanted adjustable ventriculoperitoneal hydrocephalus shunt valves. These models were created with physician feedback and evaluated by 13 health care professionals. Based on tactile evaluations and qualitative feedback, three models were selected as most clinically relevant. An optical measurement and valve setting method were developed to characterize programming tool movements during valve programming procedure and usability testing that evaluated over 50 health care professionals that program hydrocephalus valves.
The skull models used a variety of synthetic tissues and thicknesses simulating a 3mm scalp thickness protruding valve model (such as younger or older patients with thin skin), a 7mm scalp thickness model (that characterizes ‘average’ patients) and a 10mm thick model (simulating post-surgical edema or tissue scarring over a valve implanted for years). Different colors of ultraviolet fluorescent invisible ink markers were used to draw dot pairs relative to the valve center. Precision drilled tool kits (precision machined pairs of holes) were developed to characterize angular offset in addition to actual offset from valve center. Models were covered in plastic wrap enabling the same models to be reused throughout usability studies minimizing error. Various cameras were used (video and still) with visible and ultraviolet light photography throughout usability testing.
The printed grid overlays were on average 0.05mm +/- 0.35 mm offset compared to the drawing; The machined toolset hole centers were on average -0.03mm +/- 0.40 mm offset compared to the drawing; The markers dots were on average 0.15mm +/- 0.30 mm offset compared to the drawing; The valve center was on average 0.43mm +/- 0.22mm offset from the printed grid overlay and had a difference in angle of -0.26 +/- 1.37 degrees from the actual position of the valve. Real time valve setting was enabled with an endoscope placed underneath the programmable valve.
It is very important to accurately characterize human factors while developing medical devices. Clinically relevant models and these measurement methods enabled characterization of programmable hydrocephalus valve tool kit usability.
- The 5th Chinese Neuro Trauma Forum - The South Neurosurgery Forum: 2013, (Guangzhou, China) Combined meeting podium presentation - Intracranial Pressure (ICP) Monitoring, Guidelines, Present and Future Patient CareGoogle Scholar
- Opal – Medical Devices Summit: 2013, (Boston, MA) Best Practice in Postmarket Surveillance: Panel discussion - Meet the regulatory requirements in safety monitoring (Track adverse events among a growing set of information, Collect current, accurate data with minimal “noise” ensuring important signals are not missed, Maintain a central repository with up-to-date, accurate and comprehensive information, Postmarket Surveillance Abroad: Effective Data Management and Regulatory ComplianceGoogle Scholar
- North American Neuromodulation Society: Edited by: Michael A Luedtke, Jeyakumar Subbaroyan, Jeffery A. Murphy Anthony DiUbaldi, J Thomas Megerian. 2012, Poster 130 - Neuromodulation of Healthy Median Nerve by Means of Stimulation with a Transdermal Amplitude Modulated Signal (TAMS)Google Scholar
- Melissa Chu, Nikiforos Kollias, Luedtke Michael: Confocal Scanning Laser Microscopy: Applications for Imaging Dynamic Processes in Skin In Vivo, Laser Scanning, Theory and Applications. Edited by: Chau-Chang Wang. 2011, ISBN: 978-953-307-205-0, InTech, Book Chapter link: http://www.intechopen.com/articles/show/title/confocal-scanning-laser-microscopy-applications-for-imaging-dynamic-processes-in-skin-in-vivoGoogle Scholar
- Fan C, Luedtke M A, et al: “Characterization and quantification of wound-induced hair follicle neogenesis using in vivo confocal scanning laser microscopy.”. Skin Research and Technology. 2011, 17 (4): 387-397. 10.1111/j.1600-0846.2011.00508.x.PubMed CentralView ArticlePubMedGoogle Scholar
- Luedtke M A, Papazoglou E, et al: “Wavelength effects on contrast observed with reflectance in vivo confocal laser scanning microscopy.”. Skin Research and Technology. 2009, 15 (4): 482-488. 10.1111/j.1600-0846.2009.00394.x.View ArticlePubMedGoogle Scholar
- Stamatas GN, Nikolovski J, Luedtke MA, Kollias N, Wiegand BC: “Infant Skin Microstructure Assessed In Vivo Differs from Adult Skin in Organization and at the Cellular Level.”. Pediatric Dermatology. 2009, 2009Google Scholar
- Spencer J M, Kurtz E S: “Approaches to Document the Efficacy and Safety of Microdermabrasion Procedure.”. Dermatologic Surgery. 2006, 32 (11): 1353-1357.PubMedGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.